M133C 3F Power & Energy Calibrator
Three phase electric power/energy calibrator with power quality functions for calibration and testing of grid measuring appliance like power meters, power transducers or power quality analysers.
 Power quality and energy functions
 0.01° phase shift accuracy
 15  1000 Hz output
 Builtin multimeter for transducer calibration
 Floating current outputs for 3wire power meters
 Voltage from Current function
M133C is electric AC/DC power and energy calibrator for calibration of power meters, power transducers,
power quality analysers and generally all kinds of power measuring devices. AC voltage, current, power
and energy functions have calibrated phase shift in frequency range from 15 Hz to 1 kHz with resolution
as good as 0.001 Hz. Simulated electric power (including power quality functions) can be set independently
in each phase of both voltage and current signals. Builtin 5½ process multimeter for transducer calibration
is standard part of every M133C calibrator.
Languages: English, Chinese (simplified)
Power quality functions
Four different distortion functions are available in M133C calibrator: Harmonic distortion with up to 50 harmonic products, interharmonic distortion up to 5000 Hz, amplitude modulation using symetrical sine or square wave envelope and finally dip/swell amplitude modulation using custom square wave envelope. Modulation frequency can be set in range from 0.001 Hz to 50 Hz, maximum modulation depth is 30 % and again, all the distortion parameters can be set independently for each phase.
Calibration versatility
M133C is more than just a sophisticated power source. Using builtin process multimeter, both transducer inputs and outputs can be handled by the M133C at the same time so you can calibrate it more easily, using no other calibration equipment. Floating current outputs can be connected directly with voltage outputs to allow for 3wire power meter calibration. One calibrator, many applications  saving your time, space and costs.
DC voltage 
3x 1  280 V ± 150 ppm / year 
AC voltage 
3x 1  600 V, 15  1000 Hz ± 120 ppm / year 
DC current 
3x 5 mA  30 A (2250 A with 15125 Current Coil) ± 175 ppm / year 
AC current 
3x 5 mA  30 A, 15  1000 Hz (2250 A with 15125 Current Coil) ± 175 ppm / year 
DC Power & Energy 
3x 5 mW  8.4 kW (3x 420 kW with 14050 Current Coils) ± 380 ppm / year 
AC Power & Energy 
3x 5 mW  18 kW, 15  1000 Hz (3x 900 kW with 14050 Current Coils) 0.01° phase shift accuracy, power factor 1 to +1 Harmonic, interharmonic, modulation/flicker, dip/swell distrortion functions ± 370 ppm / year 
Process Multimeter
DC voltage 
0  12 V ± 100 ppm / year 
DC current 
0  25 mA ± 100 ppm / year 
Frequency 
1 Hz  15 kHz ± 50 ppm / year 
High Current Adapter 13301
joins all 3 current outputs to increase output range to 90 A


Current Coil 0950
multiplication factors 10x, 25x and 50x, 1500 A maximumgo to product page 

Current Coil 15125
x25 multiplier, 3000 A maximum, active coolinggo to product page 