H1,25.06.2021 14:32:51,1838,!Version: 1.00|Author: Meatest|E-mail: meatest@meatest.cz
OC'0,19999,3,3,0.5,1'
<
J,V, COM
F1
P,0.05,,,1,6000
R0=<=0.6
P,0.1,,,1,6000
K,,,,-1,-1
R0=<=6
R0=<=60
R0=<=600
R0=<=1000
P,0.05,,,1,1000
F3
P,1,,,4,6000
R0.018=<=0.6
K,,,,-1,-1
E15=<45
E45=<=65
P,0.7,,,4,6000
E65<=1000
E1000<=5000
P,2,,,4,6000
E5000<=20000
P,2,,,20,6000
R0.18=<=6
E15=<45
E45=<=65
P,0.7,,,4,6000
E65<=1000
E1000<=5000
P,2,,,4,6000
E5000<=20000
P,2,,,20,6000
R1.8=<=60
E15=<45
E45=<=65
P,0.7,,,2,6000
E65<=1000
E1000<=2500
P,2,,,4,6000
R18=<=600
E15=<45
E45=<=65
P,0.7,,,2,6000
E65<=1000
E1000<=2500
P,2,,,4,6000
R30=<=1000
P,1,,,4,1000
E15=<45
E45=<=65
P,0.7,,,2,1000
E65<=1000
F5
J,mA, COM
P,0.2,,,4,6000
R0=<=0.0006
K,,,,-2,-2
R0=<=0.006
P,0.2,,,2,6000
K,,,,-2,-2
R0=<=0.06
K,,,,-1,-1
R0=<=0.4
K,,,,-1,-1
P,0.2,,,2,4000
R0=<=6
J,A, COM
R0=<=10
J,A, COM
P,0.2,,,2,1000
F6
J,mA, COM
P,1,,,2,6000
R0.000018=<=0.0006
K,,,,-2,-2
E45=<=2000
R0.00018=<=0.006
K,,,,-2,-2
E45=<=2000
R0.0018=<=0.06
K,,,,-1,-1
E45=<=2000
R0.02=<=0.4
P,1,,,2,4000
K,,,,-1,-1
E45=<=2000
R0.18=<=6
J,A, COM
E45=<=2000
R1.5=<=10
P,1,,,2,1000
J,A, COM
E45=<=2000
F7
K,,,,1,1
P,0.2,,,1,6000
R0=<=600
P,0.2,,,2,6000
K,,,,0,0
R0=<=6000
R0=<=60000
R0=<=600000
R0=<=6000000
K,,,,2,2
R0=<=30000000
K,,,,2,2
P,1,,,1,3000
R0=<=50000000
K,,,,2,2
P,1.5,,,1,5000
F11
K,,,,-2,-2
P,1,,,2,1000
R0=<=0.00000001
K,,,,-3,-3
R0=<=0.0000001
K,,,,-3,-3
R0=<=0.000001
R0=<=0.00001
R0=<=0.0001
R0=<0.01
P,1,,,2,10000
F13
P,0.005,,,1,20000
R0.5=<200
R0.5=<2000
R0.5=<20000
K,,,,1,1
R0.5=<200000
K,,,,1,1
F48
P,1,,,10,10900
R-200=<=1090
F47
P,1,,,10,10900
R-200=<=1090
#F1:*VDC-2W,V,a,,,,m,%
#F3:*VAC-2W,V,a,Hz,Frequency,,,%
#F5:*IDC,A,a,,,,m,%
#F6:*IAC,A,a,Hz,Frequency,,,%
#F7:*RDC-2W,Ohm,a,,,,,%
#F11:*C-2W,F,a,,,,,%
#F13:*FREQ1,Hz,a,,,,,%
#F48:*TC K S68,°C,a,°C,Remote junction
#F47:*TC K S90,°C,a,°C,Remote junction